**M100 specification**

**Environment **

Working temperature 0 to +60 °C

Functional temperature -25 to +70 °C

Storage temperature -55 to +85 °C

Temperature coefficient 0.02% per °C (100ppm/°C)

Relative humidity 95% non condensing

Class of climate HSE complying with DIN40040-3 complying with VDE/VDI3540

**Accuracy **

Class ±0.2 % complying with IEC688

Calibration temperature 23°C

Temperature coefficient 0.01% / °C (100 ppm / °C)

Stability 0.05 % per annum non cumulative

Warm up time <15 min

**Outputs**

Rated value

See individual product pages.

Load resistance (current output)

1mA <15 kOhm

5mA <3 kOhm

10mA <1.5 kOhm

20mA < 0.75kOhm

4-20mA < 0.75kOhm

Load resistance (voltage output)

1V, 5V & 10V >1 kOhm

(M100-VA1,VA3) 1V, 5V & 10V > 50kOhm

Load influence < 0.1 %

Ripple < 0.5% peak-peak at full load

Response time < 200 ms for 0-99 % at full load

Overload < 2 x rated value at full load

No load voltage < 27 V

**Approvals**

U.L. Approval File No E157034

**Insulation***Test Voltage:*

4kV RMS 50Hz 1min. between Input/Case/Auxiliary/Output *Impulse Test:*

EMC 5kV transient complying with IEC801/EN55020*HF interference:*

EHF 2.5kV 1MHz complying with IEC255-4*Protection:*

Class II complying with IEC348 BS4753/DIN57411/VDE 0411

**Enclosure**

Fixing - Snap on to DIN rail 35 x 7.5 mm complying with DIN-EN50022 BS5584

Mounting - Any position

Enclosure Code - Case IP50/terminals IP30 complying with IEC529 BS5490 DIN40050

**Applied Standards**

General IEC688/BS6253/VDE/VDI2192

Safety EN61010-1 DIN57411/VDE0411 ANSI C37

Surge withstand IEC801/EN55020 ANSI C37-90a

Radio screening RFI degree N complying with VDE0875

EMC Emissions EN61326-1

Immunity EN61326-2

**M100 Active Power Transducer**

**Selection Guide**

M100-WA1 Single phase

M100-WA2 3 phase 3 wire balanced load

M100-WA3 3 phase 4 wire balanced load

M100-WA4 3 phase 3 wire unbalanced load

M100-WA5 3 phase 4 wire unbalanced load

M100-WA6 3 phase 3 wire balanced load externally connected reverse CTs

M100-WA7 3 phase 3 wire balanced load internally reversed CTs

**Typical Applications**

The M100-WA series measure active power in single, three phase three or four wire balanced and unbalanced systems.

Using the time division multiplier circuit means that they can be used over a wide range of input waveforms. The D.C. Output signal is directly proportional to the instantaneous power being measured.

Typical applications include the measurement of power in switchboards, power stations, generating sets etc.

The high isolation of 4kV as with all the M100 series, allow these watt transducers to be connected to a variety of measuring and control devices and systems, such as analogue meters, PLC, computers, data loggers, digital instruments and telemetry systems.

Both auxiliary powered and self powered versions of each type are available, it is recommended to use an auxiliary powered version if the system being measured has voltage variations in excess of ± 20%.

**Ordering Information**

Product Code I/P In Un O/P Range Aux Freq Opt.

M100-WA5 800/5A 230v 0-20mA 600kW 230v 50Hz

**Options **

1. Non standard inputs / outputs only as far as technically acceptable.

2. A.C. Auxiliary in range 57.7 to 450 volts

3. Calibration at nominal Hz 35.....450Hz

4. Calibration at temperature other than 23°C

**Technical Specification**

Rated value In 1 or 5 Amp C.T. connected

0.5-10 Amp direct connected

Rated value Un 57.8 < 600 volt

Power consumption <1 VA voltage

<0.2 VA current

Working range 0-125% Un auxiliary powered

75-125% Un self powered

0-150% In

Rated Frequency 50 / 60 / 400 Hz

Frequency influence 0.005 % / Hz

Overload continuous 4 x In 1.5 x Un

Overload for 1 sec. 50 x In 2 x Un

Rated value mA 0-1/5/10/20 & 4-20mA

Rated Value Volts 0-5 / 10 & 1-5 V

Zero ± 2%

Span ± 10%

A.C. Voltage 115 / 230 / 400 V (± 25% / 45-65 Hz / < 2VA)

D.C. Voltage 24 / 48 / 110 V (± 20% / galvanically isolated / <3 W)

M100-WA1,2,3,6,7 Approx. 0.6kg. 100mm case

M100-WA4,5 Approx. 0.8kg. 100mm case

**M100 Reactive Power Transducer**

**Selection Guide**

M100-XA1 Single phase

M100-XA2 3 phase 3 wire balanced load

M100-XA3 3 phase 4 wire balanced load

M100-XA4 3 phase 3 wire unbalanced load

M100-XA5 3 phase 4 wire unbalanced load

M100-XA6 3 phase 3 wire unbalanced load

M100-XA7 3 phase 3 wire balanced load internally reversed CTs

**Typical Applications**

The M100-XA series measure reactive power in single, three phase three or four wire balanced and unbalanced systems.

Using the time division multiplier circuit means that they can be used over a wide range of input waveforms.

The D.C. Output signal is directly proportional to the instantaneous reactive power being measured.

Typical applications include the measurement of reactive power in switchboards, power stations and generating sets etc.

The high isolation of 4kV as with all the M100 series, allows these VAr transducers to be connected to a variety of measuring and control devices and systems, such as analogue meters, PLC, computers, data loggers, digital instruments and telemetry systems.

Both auxiliary powered and self powered versions of each type are available.

It is recommended to use an auxiliary powered version if the system being measured has voltage variations in excess of ± 20%.

**Ordering Information**

Product code I/P In Un O/P Range Aux. Freq. Opt.

M100-XA4 400/5 400 0-20mA 300kVAr 120 60Hz

**Options **

1. Non standard inputs / outputs only as far as technically acceptable.

2. A.C. Auxiliary in range 57.7 to 450 volts

3. Calibration at nominal Hz 35.....450Hz

4. Calibration at temperature other than 23°C

**Technical Specification**

Rated value In 1 or 5 Amp C.T. connected

0.5-10 Amp direct connected

Rated value Un 57.8 < 600 volt

Power consumption <1 VA voltage

<0.2 VA current

Working range 0-125% Un auxiliary powered

75-125% Un self powered

0-150% In

Rated Frequency 50 / 60 / 400 Hz

Overload continuous 4 x In 1.5 x Un

Overload for 1 sec. 50 x In 2 x Un

Rated value mA 0-1/5/10/20 & 4-20mA

Rated Value Volts 0-5 / 10 & 1-5 V

Zero ± 2%

Span ± 10%

A.C. Voltage 115 / 230 / 400 V (± 25% / 45-65 Hz / < 2VA)

D.C. Voltage 24 / 48 / 110 V (± 20% / galvanically isolated / <3W)

M100-XA1,2,3,6,7 Approx. 0.6kg. 100mm case

M100-XA4,5 Approx. 0.8kg 100mm case

**M100 Phase Angle Transducer**

**Selection Guide**

M100-PA1 Single phase 4 quadrants

M100-PA2 3 phase 3 or 4 wire balanced 2 quadrants

M100-PA3 3 phase 3 or 4 wire balanced 4 quadrants

M100-PV1 Single phase 4 quadrants phase angle between two voltages

**Typical Applications**

The M100-PA series of phase angle transducers measure the phase angle between current and voltage.

They can be used on single and three phase three or four wire balanced systems and are ideal for optimising power factor correction.

The M100-PV2 measures the phase angle between two voltage supplies and provides a D.C. output signal proportional to the phase angle between the voltages.

**Ordering Information**

Product code I/P In Un O/P Range Aux. Freq.Opt.

M100-PA2 5Amp 400V ± 45° 120V 60Hz

**Options **

1. Non standard inputs / outputs only as far as technically acceptable.

2. A.C. Auxiliary in range 57.7 to 450 volts

3. Calibration at nominal Hz 35.....450Hz

4. Calibration at temperature other than 23°C

**Technical Specification**

Rated value In 1 or 5 Amp C.T. connected

0.5-10 Amp direct connected

Rated value Un 57.8 < 600 volt

Power consumption <1 VA voltage (aux powered)

<2.5 VA voltage (self powered)

<0.2 VA current

Working range 15-125% Un auxiliary powered

75-125% Un self powered

10-150% In

Measuring range ± 45 / 60 / 90 / 180° M100-PA1

± 45 / 60° M100-PA2 ± 90 / 180° M100-PA3

Rated Frequency 50 / 60 / 400 Hz

Frequency influence 0.005 % / Hz

Overload continuous 4 x In 1.5 x Un

Overload for 1 sec. 50 x In 2 x Un

± 1 degree

Rated value mA 0-1/5/10/20 & 4-20mA

Rated Value Volts 0-5 / 10 & 1-5 V

Zero ± 2%

Span ± 10%

A.C. Voltage 115 / 230 / 400 V (± 25% / 45-65 Hz / < 2VA)

D.C. Voltage 24 / 48 / 110 V (± 20% galvanically isolated / <3W)

Approx. 0.6 kg. 100mm case